The Calibration Kit has been created to provide all
the necessary components to enable customers to
verify the accuracy of the readings on the Buckleys or
Roxby Bathycorrometers so that in-house
calibration certificates can be issued.
The process requires two areas to be checked. The
electronics should be tested using the BCM Checker.
Readings on the Bathycorrometer should be 0.000
with the checker switched off and 1.990 with the
Providing the readings are correct, the Silver/Silver
Chloride screw-in half cell should be checked by
using K-Series Reference Cells in a standard salt
water solution. To confirm the readings, three cells
should be used and the Sea Water corrosion test
mixture should be made up to a 3% solution.